# Area Under Curve Measurement

**Area Under Curve Measurement**, a forum discussion on Cleverscope Mixed Signal USB Oscilloscopes. Join us for more discussions on

*Area Under Curve Measurement*on our Interesting Questions forum.

Back to Forum Index : Back to Interesting Questions |

**rhcarter**

20 Nov 2007

Posts: 99

You can measure the area under a curve using the integral process in the Maths Equation builder.

Open the Maths equation builder, and type 'a' in the function, set the process to Integral, and set the destination to 'a.' Click on 'Apply Equations'. Open the Maths window. The window will show the time integral of the 'a' channel. If this is a very short period (eg a few microseconds), the integral will be correspondingly small (eg an integral of 1V over 1us will yield 1Vus). If you want the numbers to be bigger, you could pre-multiply channel 'a' by an appropriate factor in the equation builder - eg a*1000000, if working in us.

Roger

Open the Maths equation builder, and type 'a' in the function, set the process to Integral, and set the destination to 'a.' Click on 'Apply Equations'. Open the Maths window. The window will show the time integral of the 'a' channel. If this is a very short period (eg a few microseconds), the integral will be correspondingly small (eg an integral of 1V over 1us will yield 1Vus). If you want the numbers to be bigger, you could pre-multiply channel 'a' by an appropriate factor in the equation builder - eg a*1000000, if working in us.

Roger

[editor:html]

**rhcarter**

29 Apr 2008

Posts: 99

We now have an article available on the resource page - http://www.cleverscope.com/resources/ called "Integration and filtering" that will give you a step by step guide on this process.

[editor:html]

Back to Forum Index : Back to Interesting Questions |