# Logarithmic Scale

**Logarithmic Scale**, a forum discussion on Cleverscope Mixed Signal USB Oscilloscopes. Join us for more discussions on

*Logarithmic Scale*on our Questions forum.

Back to Forum Index : Back to Questions |

**Ash**

28 Jun 2012

Posts: 2

Does anyone know how I can get a CS328A to display a pulse train with a logarithmic vertical scale rather than the more Common linear voltage scale. I want to display time on the X axis and a Logarithmic voltage scale on the Y axis

**bartschroder**

2 Jul 2012

Posts: 471

We don't have a log amplitude axis available (ie one that went 1 10 100 1000 up each division).

However, you can do maths to display a large dynamic range.

You can multiply the logarithm as needed.

You can use the Analog Names and Units to correctly describe the units being used.

As an example, you wish to display log base 10 of the pulse train height.

In the Maths Equation Builder (View/Maths Equation Builder) write the equation

log(a) -> b

This will send the log of Channel A to the B channel.

Open the Maths Display (View/Maths Display), and you will see the logarithm of the pulse train amplitude in the B channel. The amplitufe axis will be the usual 0 1 2 3 4 but I guess you will have to infer 1 10 100 1000 10000

Here is an image of a 4000V offset sinewave, and the conversion.

However, you can do maths to display a large dynamic range.

You can multiply the logarithm as needed.

You can use the Analog Names and Units to correctly describe the units being used.

As an example, you wish to display log base 10 of the pulse train height.

In the Maths Equation Builder (View/Maths Equation Builder) write the equation

log(a) -> b

This will send the log of Channel A to the B channel.

Open the Maths Display (View/Maths Display), and you will see the logarithm of the pulse train amplitude in the B channel. The amplitufe axis will be the usual 0 1 2 3 4 but I guess you will have to infer 1 10 100 1000 10000

Here is an image of a 4000V offset sinewave, and the conversion.

**Ash**

7 Jul 2012

Posts: 2

Bart thanks for that most helpful, its doing what I wanted to now, excellent.

Back to Forum Index : Back to Questions |